Sign up for the QMED & MD+DI Daily newsletter.
Endotec: Corrections and Removals
FDA’s inspection also revealed violations of the reports of corrections and removals regulation, including the following:■ Failure to submit a written report to FDA of any correction or removal of a device initiated by a manufacturer to reduce a risk to health posed by a device, within 10 days of initiating such correction or removal.
James G. Dickinson
February 1, 2010
1 Min Read
? Failure to include in the report of corrections and removals to FDA information on the device’s model, catalog, or code number and its manufacturing lot or serial number or other identification number.
You May Also Like
Masimo Takes Apple's Hail Mary Pass in StrideFeb 29, 2024|3 Min Read
Top 40 Medical Device CompaniesFeb 29, 2024|2 Min Read
2024 Planning Guide for Medical Manufacturing AutomationFeb 29, 2024|1 Min Read
Adverse Events Reported for Hologic BioZorb Marker, BioZorb LP MarkerFeb 28, 2024|2 Min Read