The world’s growing population is spurring the demand for increased access to advanced healthcare treatments. According to the United Nations Environment Programme (UNEP), the global population will surpass 9 billion by 2050, nearly 90% of which will be living in developing countries (Figure 1).1 With an aging and growing world population, there is expected to be a growing demand for healthcare products and services that use components made from platinum, other platinum group metals (PGMs), and their alloys. Increasing access to healthcare and advanced medical treatments in developing countries means platinum will play a key role in improving the quality of life of people around the world.
Among the implications of this growth trend is increased use of platinum in medical technology. Since the early 1970s, platinum has been used internationally in a variety of medical devices to treat ailments such as heart disease, stroke, neurological disorders, chronic pain, and other life-threatening conditions. The metal is used to make essential components for pacemakers, implantable defibrillators, catheters, stents, and neuromodulation devices among others. The properties of platinum that make it attractive for such applications include its biocompatibility, inertness within the body, durability, electrical conductivity, and radiopacity. The material can be used to manufacture components in a variety of forms, from rod, wire, and ribbon to sheet and foil, as well as high-precision micromachined parts. In 2010, it was estimated that 175,000 oz. of platinum was used in biomedical devices. Approximately 80% was usedin established technologies, such as guidewires and cardiac rhythm devices, while the remaining 20% was used in newer technologies, such as neuromodulation devices and stents.2
The Advantages of Platinum for Biomedical Uses
The chemical, physical, and mechanical properties of platinum and its alloys make the materials uniquely suitable for a variety of medical applications. Agnew et al.,3
and Brummer et al.,4
carried out studies that confirmed the low corrosivity, high biocompatibility, and good mechanical resistance of platinum and platinum alloys used for medical applications.
Platinum’s biocompatibility makes it ideal for temporary and permanent implantation in the body, which is exploited in a variety of treatments. As a metal, it can be fabricated into tiny, complex shapes, and it has some important properties not shared by base metals. It is inert, so it does not corrode inside the body, unlike metals such as nickel and copper that can sometimes cause allergic reactions. Modern, minimally invasive medical techniques often use electricity to diagnose and treat patients’ illnesses, and platinum’s conductivity makes it an ideal electrode material. It is also radiopaque, so it is clearly visible in x-ray images, enabling doctors to monitor the position of the device during treatment. Examples of areas where PGMs are used in medical devices, together with some of the manufacturers currently active in the medical device market, are shown in Table I.
|Medical device markets
||Examples of application areas
||Major medical device companies
||Arthroscopic; ophthalmology; endo-laparoscopic;
|Boston Scientific; Johnson & Johnson; Stryker; Tyco
||Pacemakers; defibrillators; hearing assist devices;
|Boston Scientific; Biotronik; Medtronic; St. Jude Medical
||Stents; angioplasty; catheter ablation; distal protection
||Boston Scientific; Abbott Vascular; Johnson & Johnson;
||Spinal fixation; hip implants; knee implants
||Biomet; Johnson & Johnson; Stryker; Zimmer
|Table I: Markets for Medical Devices and the Major Device Companies
For more than 40 years, platinum alloys have been employed extensively in treatments for coronary artery disease, such as balloon angioplasty and stenting, where inertness and visibility under x-ray are crucial. In the field of cardiac rhythm disorders, platinum’s durability, inertness, and electrical conductivity make it an ideal electrode material for devices such as pacemakers, implantable defibrillators, and electrophysiology catheters. More recently, its unique properties have been exploited in neuromodulation devices (including brain pacemakers, used to treat some movement disorders, and cochlear implants, to restore hearing) and in coils and catheters for the treatment of brain aneurysms.
Platinum in Biomedical Applications
Devices for Cardiac Rhythm Management. Abnormalities of the heart’s rhythm are common, often debilitating, and sometimes fatal. For example, bradycardia is a condition in which the heart’s “natural pacemaker” is too slow, resulting in fatigue, dizziness, and fainting. Other patients may be at risk of sudden cardiac death, a condition in which the heart’s lower chambers (the ventricles) fibrillate, or pulse in a rapid and uncoordinated manner. This prevents the heart from pumping blood and leads rapidly to death, unless the victim receives cardioversion (a strong electric shock to the heart that restores normal rhythm).
||Types of component
||Rod for manufacture of machine components
||Diameters from 0.001 in. (0.0254 mm) to 0.125 in. (3.175 mm); Cut lengths from 0.02 in. (0.508 mm)
|Balloon catheters; stent delivery; stimulation leads
||Guidewires; feed through wires; tip coils
|Table II: Specifications and applications of platinum and platinum alloy rod, wire, and ribbon components.
These and other cardiac rhythm disorders can now be managed successfully using implanted devices such as artificial pacemakers and implantable cardioverter defibrillators (ICDs). These consist of a pulse generator, a small box containing a battery and an electronic control system that is implanted in the chest wall, and one or more leads that run through a large vein into the heart itself. The electrodes on these leads deliver electrical impulses to the heart muscle—in the case of a pacemaker, these ensure that the heart beats regularly and at an appropriate pace, while in the case of an ICD, a much stronger electrical shock is delivered as soon as the device detects a dangerously irregular heartbeat. Each lead typically has two or more electrodes made of a platinum-iridium alloy, while platinum components are also used to connect the pulse generator to the lead (Figure 2).
Catheters and Stents. Catheters are flexible tubes that are introduced into the body to help diagnose or treat illnesses such as heart disease. The doctor can perform delicate procedures without requiring the patient to undergo invasive surgical treatment, improving recovery time and minimizing the risk of complications. Many catheters incorporate platinum components: marker bands and guidewires that help the surgeon guide the catheter to the treatment site, or electrodes that are used to diagnose and treat some cardiac rhythm disorders (arrhythmias).
One of the most common coronary complaints in the developed world is atherosclerosis. Fatty deposits harden the artery walls; this can lead to angina and ultimately to a heart attack. Blockages in the coronary arteries are often treated using a procedure called percutaneous transluminal coronary angioplasty (PTCA), also known as balloon angioplasty.This treatment uses a catheter with a tiny balloon attached at the end; this is guided to the treatment site, then inflated, crushing the fatty deposits and clearing the artery. Afterward, a small tubular device called a stent (Figure 3) is usually inserted to keep the newly cleared artery open.
||Types of component
||Radiopaque marker bands
||Inside diameter 0.0045 in. (0.1143 mm) to 0.250 in. (6.35 mm), (tolerance:
± 0.0005 in. (0.0127 mm));
Wall thickness 0.001 in. (0.0254 mm) to 0.005 in. (0.127 mm), (tolerance:
± 0.0005 in. (0.0127 mm));
Length 0.015 in. (0.381 mm) to 0.200 in. (5.08 mm), (tolerance: ± 0.003 in.
|Table III: Specifications and applications of platinum, palladium, gold, and precious metal alloy thin-walled tube components.
||Types of component
||Electrodes; machine components; tip coils
||Thickness from 0.0007 in. (0.018 mm); Width from 1.0 in. (25.4 mm) to 3.75 in. (95.3 mm)
||Imaging filters (rhodium foils)
|Table IV: Specifications and applications of platinum, platinum alloy and rhodium sheet and foil components.
The advent of the implantable metal stent, which can prop the artery open after angioplasty, reduced the occurrence of restenosis (renarrowing of the artery) by more than 25%. In 2003, FDA approved the first drug-eluting stent for use within the United States.This type of stent is aimed at further lowering the rate of restenosis following angioplasty procedures.
Platinum’s role in PTCA is to help ensure that the balloon is correctly located. The guidewire that the surgeon uses to direct the balloon to the treatment site is made of base metal for most of its length but has a coiled platinum-tungsten wire at its tip. This makes it easier to steer and ensures that it is visible under x-ray. Platinum is also used in marker bands, tiny metal rings placed on either side of the balloon to keep track of its position in the body.
Stents are usually made of base metals (typically stainless steel or cobalt-chromium). In 2009, however, Boston Scientific introduced a cardiac stent made of a platinum chromium alloy. According to Boston Scientific, the platinum alloy instills superior levels of strength, flexibility, and radiopacity. This stent has been approved in Europe, and has re cently received approval from FDA. It is described in the sidebar accompanying this article on page 40.
Catheters containing platinum components are also used to detect and treat some types of cardiac arrhythmia.Devices called electrophysiology catheters that contain platinum electrodes are used to map the electrical pathways of the heart so that the appropriate treatment—such as a pacemaker—can be prescribed.
Other catheters with platinum electrodes are used for a minimally-invasive heart treatment known as radio-frequency (RF) ablation. Arrhythmias are often caused by abnormalities in the conduction of electricity within the heart. It is often possible to cauterize part of the heart muscle to restore normal heart rhythm. For example, ablation is increasingly used to treat a very common heart problem called atrial fibrillation, in which the upper chamber of the heart (the atrium) quivers rapidly and erratically. Using a catheter equipped with platinum-iridium electrodes, the surgeon ablates, or makes small burns to, the heart tissue, causing scarring, which in turn blocks the superfluous electrical impulses that trigger the fibrillation.
Neuromodulation Devices. Neuromodulation devices deliver electrical impulses to nerves and even directly to the brain, treating disorders as varied as deafness, incontinence, chronic pain, and Parkinson’s disease. Many of these devices are based on an extension of heart pacemaker technology, and they are sometimes referred to as brain pacemakers. Like heart pacemakers, they have platinum-iridium electrodes and may also incorporate platinum components in the pulse generator.
There are a number of different types of neurostimulation, depending on the condition being treated. Spinal-cord stimulation (the most common neuromodulation therapy) is used to treat severe chronic pain, often in patients who have already had spinal surgery. Small platinum electrodes are placed in the epidural space (the outer part of the spinal canal) and connected to an implanted pulse generator. The patient can turn the stimulation off and on, and adjust its intensity.
In deep brain stimulation (DBS),the electrodes are placed in the brain itself. DBS may be used to treat pain and movement disorders such as Parkinson’s disease. It is also being investigated as a potential treatment for a range of other illnesses, including epilepsy and depression. Epileptic patients can also be treated using a vagus nerve stimulation device (the vagus nerve is situated in the neck).
A cochlear implant is used to restore hearing to people with moderate to profound hearing loss (many patients receive two implants, one in each ear). A typical device consists of a speech processor and coil, which are worn externally behind the ear; an implanted device just under the skin behind the ear; and a platinum electrode array that is positioned in the cochlea (the sense organ that converts sound into nerve impulses to the brain). The speech processor captures sound and converts it to digital information that is transmitted via the coil to the implant. This in turn converts the digital signal into electrical impulses that are sent to the electrode array in the cochlea, where they stimulate the hearing nerve. These impulses are interpreted by the brain as sound. It is believed that around 200,000 people worldwide have received one or more cochlear implants.
At present, neuromodulation is expensive and only available at a small number of specialist centers; even in developed countries, only a small proportion of potentially eligible patients receive this treatment. However, neuromodulation can be used to help patients with common and sometimes difficult-to-treat conditions (such as chronic pain, epilepsy, and migraine). Its use might therefore be expected to increase significantly in coming years as new indications for these therapies are established.
Other Implants. Platinum’s biocompatibility makes it well suited for temporary and permanent implantation in the body, a quality exploited in a variety of treatments in addition to the heart implants already discussed. Irradiated iridium wire sheathed in platinum can be implanted into the body to deliver doses of radiation for cancer therapy. This treatment takes advantage of platinum’s radiopacity to shield healthy tissues from the radiation, while the exposed iridium tip of the wire irradiates the tumor. Although this procedure is gradually being replaced by other forms of radio- and chemotherapy, it remains a useful weapon in the battle against cancer.
A more recent development is the use of coils made of platinum wire to treat aneurysms, which are balloonings in blood vessels caused by weaknesses in the vessel walls. If the patient’ s blood pressure rises, the vessel may rupture, causing a hemorrhage. Although this can occur anywhere in the body, platinum is mainly used to treat aneurysms in the brain, where surgery is difficult and fraught with risk. Platinum is used because it is inert, easy to shape, and radiopaque.
This treatment was first introduced in the late 1980s when a doctor and inventor, Guido Guglielmi,developed a detachable platinum coil that could be used to treat brain aneurysms. Coils are delivered to the site of the aneurysm by microcatheter, then detached using an electrolytic detachment process; once in place, the coils help coagulate the blood around the weak vessel wall, forming a permanent seal (Figure 4). Depending on the size of the aneurysm, between one and 30 coils are used; these are left inside the patient indefinitely. The Guglielmi Detachable Coil (GDC) device was approved in Europe in 1992 and in the United States in 1995. By 2009, it was estimated that this and subsequent generations of platinum coil technology were being used in 30–40% of patients treated for brain aneurysms in the United States.
The Manufacture of Platinum Biomedical Components
There are many technologies used to produce PGM components for biomedical applications, ranging from rod, wire, ribbon, and tube drawing, to sheet and foil manufacture and highly precise Swiss-type screw machining, or micromachining.
Rod and wire are manufactured in diameters ranging from 0.001 in. to 0.125 in. Dimensional consistency is assured by laser measurement. Rod products are used as the starting material for a variety of machine components, with most of the PGM parts being used in pacemaker, defibrillator, and other electrical stimulation products. Wire products are used primarily in three applications:
Platinum-tungsten and platinum-nickel fine wires are used on balloon catheters as guidewires for positioning the catheter in exactly the right location.
Other PGM wires are used as microcoils for neurovascular devices such as treatments for brain aneurysms.
Platinum-iridium wires are used as feed-through wires or connector wires that connect the pacemaker lead to the pulse generator.
Ribbon is manufactured in the form of continuous strips of rolled wire in a variety of platinum alloys. Ribbon is often used in place of round wire to produce coils with minimum outside diameter and is generally used for guidewire and microcoil applications. Ribbon is sometimes preferred over wire because wire can be harder to coil. It can also be used for markers instead of traditional cut tubing. Table II shows some typical specifications and applications for PGM rod, wire, and ribbon.
Fine diameter platinum, platinum-iridium, and platinum-tungsten tubing (0.125 in. internal diameter and below) cut to specific lengths is used for markers or electrodes on angioplasty, electrophysiology and neurological catheter devices, aneurism tip coils, feed-through wires used to connect the pacing lead to the pulse generator (also known as the can) that houses the hybrid microelectronics and the battery, and pacemakers. Some applications of thin-walled precious metal tubing are shown in Table III.
Sheet and foil are mainly made from pure platinum, platinum-iridium alloys, or rhodium. It can be shaped, formed, and rolled to a variety of dimensions. Sheet or foil can be cut, formed, and placed on a catheter for marking in a way similar to ribbon. Rhodium foil is used exclusively as a filter inside x-ray mammography equipment to enhance the viewing image. Table IV shows some examples of applications of PGM sheet and foil.
Micromachined parts are complex and very small—some are only 0.006 in. in diameter and barely visible to the naked eye. Fabrication must be extremely precise to maintain the necessary quality and dimensional tolerances, which can be as low as ±0.0002 in. Highly specialized equipment and techniques such as computer numerical-controlled Swiss screw machines and electrical discharge machining must be used (Figure 6). The automated high-production Swiss screw machines are used to fabricate the main components, and EDM is used to achieve the fine details required for many platinum parts.
Specialty metal micromachined parts (0.8 in. diameter and smaller) are made from a variety of materials, including pure platinum, platinum-iridium alloys, and gold, as well as nonprecious metals and alloys such as stainless steel, titanium, MP35N cobalt-nickel-chromium-molybdenum alloy, Elgiloy cobalt-chromium-nickel alloy, Kovar iron-nickel-cobalt alloy, and materials such as Vespel, Delrin, and Teflon (see Table V for examples). These products serve device applications such as coronary stents, pacemaker and defibrillator pulse generator and lead components, heart-valve splices, endoscopic catheters, blood-gas analysers, kidney dialysis, and other medical devices and related equipment.
||Other materials, metals, and alloys
||Platinum; platinum alloys; palladium; palladium alloys
||Nitinol; stainless steel; MP35N; Haynes alloy 25
||Platinum; platinum alloys
||Stainless steel 303/304/316; polymers
||Platinum; platinum alloys
||Titanium; titanium alloys; stainless steel; ceramics
||Platinum; platinum alloys; karat golds
||Platinum; platinum alloys
||Stainless steel; nitinol
|Precision pins, tips and rollers
||Platinum; platinum alloys; silver
|Bushings, shafts, shims and spacers
||Platinum; platinum alloys
|Precision fixtures and assembly tools
||Platinum; platinum alloys; Biomed series palladiumrhenium
|Brass; copper; Kovar
|Table V: Applications and materials for precision-micromachined components.
Parts made from PGMs are often complemented with a coating technology. Precious metal powders, titanium nitride, or iridium oxide are applied to create a more porous surface structure. The creation of a porous coating reduces the electrical impedance from the lead to the battery and allows for a good electrical connection, while reducing the energy needed to run the battery. This helps the battery last longer. Most pacing lead systems manufactured today have some form of porous surface. The end-use applications for coated PGM parts are the same as described above for uncoated parts.
For more than 40 years, platinum and its alloys have been used in a wide range of medical treatments, including devices such as coronary and peripheral catheters, heart pacemakers, and defibrillators. Newer technologies such as neuromodulation devices and stents also rely on the biocompatibility, durability, conductivity, and radiopacity of platinum to make key components in a variety of forms. Medical device manufacturers continue to invest in new technologies to satisfy the need for advanced medical treatments in both the developed world and, increasingly, the developing world. Platinum, the other PGMs, and their alloys will inevitably play a vital part in these developments.
2. J Butler, “Platinum 2010 Interim Review,” Johnson Matthey, Royston, UK, 2010, p. 21–22.
3. WF Agnew, TGH Yuen, DB McCreery and LA Bullara, Experimental Neurology, 1986, 92, (1), 162.
4. SB Brummer and MJ Turner, IEEE Transactions on Biomedical Engineering, 1977, BME-24, (5), 440.
This text was adapted from an article titled “A Healthy Future: Platinum in Medical Applications” by A. Cowley and B. Woodward in Platinum Metals Review, 2011, 55, (2), 98–107; doi:10.1595/147106711X566816.
1. More information on this subject can be found in the sources below:
2. “Biomaterials Science: An Introduction to Materials in Medicine”, 2nd Edn., eds. B. Ratner, A. Hoffman, F. Schoen, and J. Lemons, Elsevier Academic Press, San Diego, CA, 2004.
3. “Materials and Coatings for Medical Devices: Cardiovascular,” ASM International, Materials Park, OH, 2009.
Brian Woodward is currently the general manager of Johnson Matthey’s Medical Products business (San Diego). Alison Cowley works as principal analyst for Johnson Matthey’s Market Research department (London).