Sign up for the QMED & MD+DI Daily newsletter.
Institute Issues Alert about Infusion Pumps
ECRI Institute, a nonprofit organization that researches the efficacy of medical devices, has issued an alert about four brands of infusion pumps from Curlin Medical. The institute says the 2000 CMS, 4000 CMS, 6000 CMS, and PainSmart pumps present a risk of medication overinfusion. When engineers at ECRI Instituteâ€™s medical device laboratory tested the pumps, they found that doors on the pumps do not fully compress tubing during operation, which may result in intermittent gravity flow, releasing medication too quickly.
May 28, 2008
1 Min Read
There may be no visible indication that a pump permits gravity flow.The institute says that volume testing is the only way to determine whether a particular pump has a door problem, and recommends that users perform such a test immediately, and every three months thereafter.
You May Also Like
Masimo Takes Apple's Hail Mary Pass in StrideFeb 29, 2024|3 Min Read
Top 40 Medical Device CompaniesFeb 29, 2024|2 Min Read
2024 Planning Guide for Medical Manufacturing AutomationFeb 29, 2024|1 Min Read
Adverse Events Reported for Hologic BioZorb Marker, BioZorb LP MarkerFeb 28, 2024|2 Min Read